Program Routine Maintenance Of The Glove Box
Etelux reminds you that in addition to the standard inspection of the glove box, the each day servicing function of the glovebox is also vital
1. In advance of and right after each individual work, first clear the setting to make the ground dry and clean up. Then cleanse the equipment, which includes the interior and outer surfaces of the glove box, the observation process, the do the job floor, and so forth., to be free of charge of debris and clean.
two. Cautiously record the h2o, oxygen articles, and gas intake of the tools in the glove box just about every working day to avoid insufficient gas force and lead to the tools to halt working.
3. Pay back attention to the volume of fuel in the performing gasoline cylinder at any time and switch it in time.
four. In circumstance of energy failure, push the "vacuum pump" button on the panel to make the vacuum pump run soon after electricity on
5. Observe: Following electric power on, if there is a trouble that it cannot commence, the possibility is that the vacuum pump energy cable is loosely related. The next purpose is that the vacuum pump’s own switch is not turned on
6. When you want to place the sample into the glove box, use a smaller changeover chamber as a great deal as feasible. Prior to opening the changeover doorway, fork out notice to no matter if the strain gauge pointer on the valve cavity is at the zero situation. If it is, you can open it directly if not, you need to have to change the swap beneath the valve cavity to the "filling" placement to enable the gasoline enter the glove box transition compartment, and then change the switch back again to the intermediate situation, then you can open the changeover compartment Place the sample in the doorway, and then shut the doorway. When placing samples, be watchful not to touch the internal wall of the transition chamber with your arms, have on gloves to choose and location the sample
Doing these routine maintenance tasks can lower the decline amount of the glove box, prolong the support lifetime of the Glove Box Accessories box, and also lower the prevalence of accidents